HomeGrown Solutions is open for submissions. Submissions are reviewed in 4 time periods: January 15th, April 15th, July 15th, and October 15th. Submissions received after each of these dates will be reviewed in the next timeframe.

SimMom Hand Pump

Submitted by: Jennifer Bartlett on January 30, 2015
HomeGrown Solution Number: 246
Identification of the Problem
Although there is the functionality to program fluid (simulated blood) flow in a Laerdal SimMom scenario, the simulationist has minimal control over the rate and timing of the fluid flow. In the face of a pump failure, an external manual pump is the most viable back-up option.
Unique Idea
As a one-time troubleshooting fix for a failed pump, we installed a hand fluid/siphon pump. Although originally intended as a temporary fix, this option provides optimal control of the rate and timing of fluid flow, so we have implemented this equipment/process on a more permanent basis. During a simulation, when students are specifically assessing for bleeding (locia), we begin pumping; we cease pumping when students are otherwise engaged, which means we are using less than half of the simulated blood we were using before this invention.
Objectives
Supplies/Ingredients
 1. 1 - hand fluid/siphon pump (with tubing attached) ($6.00) 2. 10 feet (more/less depending on your set-up) of ½ inch OSD air tubing ($5.00) 3. Industrial waterproof tape (duct tape) ($3.00) 4.$3.00 5. 10 feet (more/less depending on your set-up) of 1-3/8 inch wire wrap ($3.00) 6. 1 - double-ended tubing connector (set of 10 for$3.00 at an automotive shop)
Steps to Creating the Solution
 1. Gather materials. 2. Place the hand pump where you would like it to be placed for pumping during the simulation (to determine air tubing length needed). 3. Attach the out-pump tubing from the hand fluid/siphon pump to the new air tubing (long enough to reach SimMom’s pelvis), and secure with a clamp (included in our hand fluid/siphon pump kit) or tape). 4. Remove the red Luer connector (and the white elbow) from the existing blood feed tube. 5. Connect the end of the new air tubing (out-pump side of the hand fluid/siphon pump) to the existing blood feed tube (without the Luer connector) using a double-ended tubing connector. 6. Secure the tubing as needed with industrial waterproof tape. 7. Place the end of the in-pump tubing in a large (half-gallon) container of simulated blood. 8. Cover the exposed tubing by wrapping dark wire wrap to disguise the fluid flow. 9. Prime the tubing by pumping. 10. Pump as needed during simulation.